Rika is a sensor manufacturer & solution provider of environmental and weather monitoring for 10+ years
Installation steps of noise and dust monitoring system
Air pollution will endanger people's health, affect the physiological functions of plants, reduce plant yields, deteriorate plant quality, affect weather and climate, and cause irreversible harm to the ecological environment. The country has always attached great importance to the prevention and monitoring of air pollution, especially the monitoring of dust pollution. Many dust spots, wide areas, and long lines have become the main causes of air pollution in central urban areas. Dust pollution mainly comes from four aspects: First, construction sites and road construction sites, most of which have not implemented dust control measures such as road hardening, vehicle washing, slag coverage, and land fencing. Second, the vehicles that transport cement, sand, gravel, and muck were not completely closed for transportation, and the spills were heavy, polluting the roads, and increasing dust on the roads. Third, the concrete mixing plant did not operate according to the standard, the materials were stacked in the open air, and the dust and road dust pollution were serious. Fourth, road construction is basically open operation, no dust prevention measures, no enclosed or semi-enclosed road fences, etc. The monitoring of these aspects is generally carried out by installing noise and dust monitors on construction sites, road construction sites, and concrete mixing stations. The installation steps of the noise and dust monitor:1. Inventory materials (take ten monitoring as an example)One louver box all-in-one transmitter, one white tray Pcs, 3 M4*10 screws, 1 white pole (default 3m length, composed of 2 sections of 1.5m), beam (1 U-bolt, 2 0.75 nuts), 2 hoop (including mounting screws) , 8 M10*20 screws) Noise and dust all-in-one machine (including 1 key), wind speed sensor, wind direction sensor, 1 one-to-three cable, 2 black brackets, 12 M4*10 screws. Second, install the sensorUse the brackets to fix the shutter box, wind speed sensor, and wind direction sensor on the beam in turn. When installing the wind direction, pay attention that the arrow on the sensor must face the north to avoid measurement Error. Third, install the beam Fix the beam with the sensor installed on the pole. 4. Fix the LED screen and electric control boxRequired accessories: hoop, screw5. Wiring There will be one in the accessories Drag the three wires, plug the three ends into the wind direction sensor, the louver box all-in-one transmitter, and the wind speed sensor in turn (the connector can be plugged into three wires without distinction), and the other end is installed at the bottom of the box. VI. Ground cage installation Fix the installed dust detector to the ground with expansion screws, and power on to view the data. Seven, install the APP and configure the parameters download and install the configuration tool, search for the wireless network and connect when the dust host is turned on. Then click on the next good configuration software, you can configure the display header, spray upper and lower limits and other parameters without an account and password. Before, my country's environmental supervision capacity building level was generally low. In terms of environmental quality monitoring, monitoring equipment is insufficient, and air quality automatic monitoring network coverage is incomplete; in terms of environmental law enforcement, grass-roots law enforcement equipment is relatively backward, and the operation system of automatic monitoring facilities for key pollution sources is incomplete; There is a large gap in the national standardization requirements, and it is difficult to meet the needs of air pollution prevention and control work. The popularization of noise and dust monitoring systems has improved this situation. The noise dust monitoring system consists of a noise dust monitoring station, a transmission system, a monitoring software cloud platform and a spray system (fog cannon), which can monitor PM2.5, PM10, ambient temperature and humidity, and wind speed and direction in the on-site environment. Noise monitoring, video monitoring and video capture of pollutants exceeding the standard (optional), toxic and hazardous gas monitoring (optional) and other functions; upload the monitored data to the data platform, which is a networked platform with an Internet architecture , It has the monitoring function of each sub-station and the alarm processing, recording, query, statistics, report output and other functions of the data. The system can also be linked with various pollution control devices to achieve the purpose of automatic control. At the same time, in order to facilitate the monitoring of relevant departments, the data monitored by the dust monitor can be directly uploaded to the government environmental monitoring platform to provide relevant basis for environmental law enforcement.
In an age when sensor solution is increasingly important, the researchers believe manufacturers should pay close attention to their results.
For more advice on tips, please visit our website Rika Sensors. Do not hesitate to contact us if you are interested.
It is never too late to have a new mindset and to get things moving in the right direction. Choose Hunan Rika Electronic Tech Co.,Ltd to be your quality provider.
sensor solution has obtained many affirmation in the market. Undoubtedlly, our customers are totally satisfied with our products.