0 ~ 0.5m…200mH2O or 0 ~ 5KPa…2MPa
The liquid(not sticky ) compatible with 316 stainless steel
0.1%FS/year typ.,0.2%FS/year max.
Sensor:316L,housing:304SS(316L is optional)
Outer material：PUR,Atmospheric pressure compensation cable,
Current output:(U*0.02)W, Voltage output:(U*0.008)W, Digital output:(U*0.015)W
Current output:≤(U-7)/0.02Ω, Voltage output:≥100kΩ
DIMENSION & FIXED BRACKET(OPTIONAL)
As shown in picture 1,when the sensor is installed in static water such as in poos,water towers,probe is immersed into the bottom and should be as far as possible away from the pump or valves.The terminal box should keep above water surface and prevent water penetrating into cables.Please make sure the airway not be blocked.
As shown in picture 2,when the sensor is installed in dynamic water such as dams or rivers,probe should be inserted into a steel pipe(inner dia is around 45mm),burrowing several holes at different height on the pipe side wall which is opposite to the water flow direction.The terminal box should keep above water surface and prevent water penetrating into cables.Please make sure the airway not be blocked.
Copyright © 2019 Hunan Rika Electronic Tech Co.,Ltd | All Rights Reserved
We are here to help you! If you close the chatbox, you will automatically receive a response from us via email. Please be sure to leave your contact details so that we can better assist